No stress rentals! We help you to choose the equipment that you need, we expedite shipment, and follow up. Hioki Power Quality Analyzer No stress rentals! Display items: Amplitude and duration of swell Measurement accuracy: Same as RMS voltage Voltage dip drop in RMS voltage Measurement method: True RMS a single cycle is calculated by overlapping each half cycle The voltage between lines is measured for three phase 3-wire lines, and phase voltage is measured for three phase 4-wire lines. Inter-harmonic voltage and current Analysis window: Rectangular Analysis orders: 0.
|Published (Last):||19 October 2013|
|PDF File Size:||3.4 Mb|
|ePub File Size:||5.99 Mb|
|Price:||Free* [*Free Regsitration Required]|
Quick Links. Download this manual See also: Instruction Manual , Manual. Table of Contents. Note that EN mode is not available in versions prior to Ver1. For operating environment, maintenance, and disposal at end of life, the same conditions apply as to the main Indicates the name of an item displayed on the screen. Indicates operation keys. Displays the item selected in the drop-down list box accompanied by an explanation. Page 7: Overview Overview 1. When EN mode is selected, the measurement function for EN is added to the normal mode, and some functions in nor- mal mode will be limited.
Page En Screen 2. In general, there are 3 screen types, and measurement in EN mode is possible only in these screens. In addition, screens for normal mode e. Page Pc Card 2. Using a PC card is recommended for further analysis after mea- surement by using the PC application software.
Page Limitations In En Mode 2. Page Default Values 2. The settings for other parameters are not reset. Page Measurement Measurement 3. By initializing the settings before starting measurement, EN measurement can be started without making further complicated settings.
Example When the start key is pressed at , the waits to start measurement until Page Common Display Areas 3. Page 19 The ratio of Good judgment in the designated period is displayed as a numeric value and bar graph. Page Overview Screen 3. The details of each screen are explained as follows.
A Freq. B Vvari. Page 21 Changing the display method Overview example Select from pull- down menu Display the data from measurement start to the present or end time Display the data for a designated period Previous recorded prior to the present or end time Display the data for a designated period Specific Confirm Cancel from a specified date Note that only periods shorter than the recorded period can be selected. Page Harmonic Screen 3. Harmonic Changing the display method Harmonic SELECT example Select from pull- Display the data from measurement start down menu to the present or end time Display the data for a designated period Previous recorded prior to the present or end time Page 25 Changing the flagging Harmonic Incl.
Flag Excl. Flag Include flagged data in the Including flagged data statistics Do not include flagged data in Excluding flagged data the statistics This is offered to see the judgment without the effect of dip, swell and interruption events which affect all parameters. Page Signaling Screen 3. Signaling The data displayed in the Signaling screen is one-day data of the specified date, because EN standards require that signal- ing be assessed terms of one-day periods. However, signaling in the Overview screen is the result of statistics from the same period as other parameters.
Page 27 Changing the flagging Signaling Incl. Page Events Screen 3. The time and level cannot be changed. Events In the normal mode, when an event occurs, an event is counted twice.
Page 29 Changing the display method Events example Select from pull- down menu Confirm Cancel Display the data from measurement start to the present or end time Display the data for a designated period Previous recorded prior to the present or end time Display the data for a designated period Specific from a specified date Page Detailed Settings Detailed Settings 4. Details for each setting are as follows: Settings1 Settings for the Measurement system.. As such, setting in the system screen of normal mode will not be possible.
Confirm Cancel You can set the optional nominal voltage within the 50 to range. Setting value Interruption Settings1 U Interrupt Set interruption value. Page 38 Short interruption time Settings1 Short int T The time for a short interruption is specified. Setting value An interruption up to this period is identified as a "short interruption. Page Threshold Setting 4. The default values are set forth by the EN standard, but they can be modified arbitrarily.
Once set, these values cannot be changed during or after measurement because the calculated data during measurement will not be stored and therefore cannot be re- analyzed. Setting value V Vari. V Vari. Page 41 Unbalance Settings2 Unbalance Set unbalance value.
This setting does not affect the measurement calcu- lations directly. The frequency range is from Hz to Hz in 5 Hz steps. Page Settings In Mode Screen 4. Set the values in normal mode as required. Page Pc Card PC application software. However, these data cannot be loaded to the unit. This data can be loaded on the main unit only. Page Specifications Specifications The specifications not indicated in this document are stated in the standard specifications for Model Setting function specifications When EN mode is selected, some functions in normal mode will be limited.
The limitations are described in below. Page 49 Event settings The following items are fixed values and cannot be changed. Page En Measurement Parameters 5. A, Freq. A, Vvari.
B Flicker severity Page En Display Specifications 5. Bar graph display The ratio of Good results in the display period is displayed as a value and a bar graph. Previous Display the data for a designated period recorded prior to the present or end time. Specific Display the data for a designated period from a specified date.
Page En Measurement Specifications 5. Page Calculation Formulas 5. Print page 1 Print document 64 pages. Cancel Delete. Sign In OR. Don't have an account? Sign up! Restore password. Upload from disk. Upload from URL.
Hioki 3196 Instruction Manual
Quick Links. Download this manual See also: Instruction Manual , Manual. Table of Contents. Note that EN mode is not available in versions prior to Ver1. For operating environment, maintenance, and disposal at end of life, the same conditions apply as to the main
Download Program for POWER QUALITY ANALYZER 3196